Air Supply (Pneumatic Requirements):
Clean, Dry Air; Minimum 10
0 psi, 10 cfm
Power Input: 100-240 VAC, 2.8-1.4 A, 50/60 Hz
Output: 24 VDC, 5 A, 120 W (for all Electronic Door Control & Components), UL and CE rated
Door Motion Transport: All Pneumatic
Door Control Signals: Electro-pneumatic, 24 VDC
Six foot light curtain on doors when closing, emergency stop push button on control panel. Pressure sensor tape to detect obstructions during opening.
User Interface: Human-Machine Interface (HMI) Touch Screen with a keyed lockout to prevent unauthorized use.
Standard User Connections: “Doors Full Open,” “Doors Full Closed,” and “Magnet in Use” signals, Fire Alarm
Standard Interlocks: One Single OR Door, One Double OR Door, MRI Magnet Mover, Other Intra-Operative Sliding Door (if Applicable)
Optional Interface/Interlock Functionality: Additional OR Door interlocks, Low Oxygen Alarm, Quench/Low Oxygen Exhaust Fans, and Air Purge Cycle Fans